On the Bench: HP 3325A Sensitized Function Generator. It had a completely sorted out bridge rectifier and a shorted transistor on the -15v rail. It is also had a broken power button, missing another button (Now on order) and was missing a “spring” on one of the buttons as well. Else, this tests out great and has a VERY clean signal and accurate output levels.
The Bad Guys….